Have a question or want to start a discussion? Post it! No Registration Necessary. Now with pictures!
- Posted on
July 19, 2011, 4:45 pm
rate this thread
Re: Newbie question about Siemens K2701 and singlemode cable
Well, I was a Newbie less than a year ago, so I'll pass on what I've
gleaned since. I've hauled a small singlemode campus network into
working for $cheap since I started, so I might have learned a few things
I don't know that meter, and Siemens appears not to be proud of it, as
you've no doubt already found (no online specs/manual in a short search.
Probably more like "they want to accelerate obsolescence, but anyway...)
If it's a power meter only, probably "yes" based on what I've been told,
and seen, with a different brand of meter.
BUT: You will need a single-mode optical source in the same
wavelength(s) as the system you are trying to measure/qualify or
whatever you are actually trying to accomplish. Depending on "for whom"
and "to what standards" you might also need to have both the meter and
the source calibrated (quite possibly costing more money than you paid
for a used meter with no manual.)
You'll also need appropriate single-mode patch cables to connect from
source to cable and from cable to meter (and also "directly to each
other" to eliminate their contribution as an initial calibration.) The
ends you connect to the cables are moved from cable to cable, but the
ends connected to the source and meter should remain connected to those
instruments throughout a measuring session, as they are much more
variable than coupling most connectors.
If you have a meter and power source in one package, it only works on
single-mode if the source is suitable for single-mode (ie, laser, not
LED, typically) - BUT: you can use a separate single-mode source, and
the meter part works just fine. I'm actually doing that, with wildly out
of date calibrations, since I can't afford expensive calibration
services and need only functional internal documentation, rather than
"traceable standardized documentation" to provide to someone else.
The effect of using a multi-mode source on single-mode fiber is much as
you might expect - very little power gets into the fiber to begin with,
as seen with a looped-back jumper cable before I got the correct (if
11-12 years overdue for recalibration to be "official") single-mode
What are you trying to do, more broadly?
Cats, coffee, chocolate...vices to live by
Please don't feed the trolls. Killfile and ignore them so they will go away.
- » NYC local event: UNIGROUP 15-JUN-2017 (Thu) Meeting: Optical Fiber...
- — Newest thread in » Fiber Optics
- » CWAers Land Major Legislative Victory For Working People In Maine [Telecom]
- — The site's Newest Thread. Posted in » General Telecommunications Forum